Apex locators, Dentaport ZX, Propex II, Pulpectomy, Radiovisiography, Working length
Citation Information :
Abdullah A, Singh N, Rathore MS, Tandon S, Rajkumar B. Comparative Evaluation of Electronic Apex Locators and Radiovisiography for Working Length Determination in Primary Teeth in vivo. Int J Clin Pediatr Dent 2016; 9 (2):118-123.
Aims: The aim of this study was to evaluate the effectiveness of two different varieties of electronic apex locators and radiovisiography (RVG) for working length determination in primary teeth.
Materials and methods: A total of 30 primary teeth indicated for pulpectomy in children aged 3 to 8 years were randomly selected and subjected to working length determination using two varieties of electronic apex locators and RVG separately. The data were then subjected to statistical analysis.
Results: A very strong correlation between electronic measurement methods and RVG length was observed.
Conclusion: Radiovisiography and apex locators are equally effective in determining working length in primary teeth.
How to cite this article: Abdullah A, Singh N, Rathore MS, Tandon S, Rajkumar B. Comparative Evaluation of Electronic Apex Locators and Radiovisiography for Working Length Determination in Primary Teeth in vivo. Int J Clin Pediatr Dent 2016;9(2):118-123.
Pratten DH, McDonald NJ. Comparison of radiographic and electronic working lengths. J Endod 1996 Apr;22(4): 173-176.
Versteeg KH, Sanderink GCH, Van Ginkel FC, Van der Stelt PF. Estimating distances on direct digital images and conventional radiographs. J Am Dent Assoc 1997 Apr;128(4):439-443.
Ellingsen MA, Hollender LG, Harrington GW. RadioVisio Graphy versus conventional radiography for detection of small instruments in endodontic length determination part II: in vivo evaluation. J Endod 1995 Oct;21(10):516-520.
Fouad AF, Reid LC. Effect of using electronic apex locators on selected endodontic treatment parameters. J Endod 2000 Jun;26(6):364-367.
Olson AK, Goerig AC, Caviato RE, Luciano J. The ability of the radiograph to determine the location of the apical foramen. Int Endod J 1991 Jan;24(1):28-35.
Gelfand M, Sunderman EJ, Goldman M. Reliability of radiographical interpretations. J Endod 1983 Feb;9(2):71-75.
Subramaniam P, Konde S, Mandanna DK. An in vitro comparison of root canal measurement in primary teeth. J Indian Soc Pedod Prev Dent 2005 Sep;23(3):124-125.
Loushine RJ, Weller N, Kimbrough WF, Potter BJ. Measurement of endodontic file lengths: calibrated versus uncalibrated digital images. J Endod 2001 Dec;27(12):779-781.
Kawauchi N, Bullen IRFR, Chinellato LEM. Evaluation of the linear measurements by conventional radiographs and indirect digital images in the endodontic treatment. J Appl Oral Sci 2004 Dec;12(4):330-336.
Sanabe ME, Basso MD, Gonçalves MA, Cordeiro RCL, Pinto LS. Digital versus conventional radiography for determination of primary incisor length. Braz J Oral Sci 2009 Apr-Jun;8(2):101-104.
Goldberg F, De Silvio AC, Manfré S. In vitro measurement accuracy of an electronic apex locator in teeth with simulated apical root resorption. J Endod 2002 Jun;28(6):461-463.
Custer LW. Exact methods of locating the apical foramen. J Natl Dent Assoc 1916; 5:815-819.
Suzuki K. Experimental study on iontophoresis. Japanese J Stomatol 1942; 16:411-429.
Goldberg F, Marroquín BB, Frajlich S. In vitro evaluation of the ability of three apex locators to determine the working length during retreatment. J Endod 2005 Sep;31(9):676-678.
Leonardo MR, Silva LA, Nelson-Filho P, Silva RA, Raffaini MS. Ex vivo evaluation of the accuracy of two electronic apex locators during root canal length determination in primary teeth. Int Endod J 2008 Apr;41(4):317-321.
Katz A, Mass E, Kaufman AY. Electronic apex locator: a useful tool for root canal treatment in the primary dentition. ASDC J Dent Child 1996 Nov-Dec;63(6):414-417.
Patiño-Marín N, Zavala-Alonso NV, Martínez-Castañón GA, Sánchez Benavides N, Villanueva-Gordillo M, Loyola- Rodríguez JP, Medina-Solís CE. Clinical evaluation of the accuracy of conventional radiography and apex locators in primary teeth. Pediatr Dent 2011 Jan-Feb;33(1):19-22.